Clear Seal Peel 9095-10101
DescriptionA clear film with good optical clarity and moderate solvent resistant properties. The film is peel-able and non-pierceable.
|9095-10101-078LR||** Std||LabRoll||1 roll||500m x 78mm|
|9095-10101-078SR||** Sterile||LabRoll||1 roll||500m x 78mm|
|9095-10101-115LR||*** VII Std||LabRoll||1 roll||350m x 115mm|
|9095-10101-115SR||*** Sterile VII||LabRoll||1 roll||350m x 115mm|
|9095-10101-078LS||* Std||LabSheet||100 sheets||125mm x 78mm|
|9095-10101-078SS||* Sterile||LabSheet||100 sheets||125mm x 78mm|
|9095-10101-078TR||Trial||LabRoll||1 roll||5m x 78mm|
|9095-10101-115TR||Trial||LabRoll||1 roll||5m x 115mm|
|9095-10101-078TS||Trial||LabSheet||5 sheets||125mm x 78mm|
Polypropylene (PP), Polyethylene (PE), Polystyrene (PS) and Cyclic Olefin Copolymer (COC) plates.
qPCR, short term compound storage.
Store in a cool place. Avoid direct exposure to sunlight. It is recommended to use the seals within three years from date of purchase. Three years when stored at 21˚C(70˚F), 50% relative humidity, out of direct sunlight, in original packaging.
Temperature range -80˚C to 80˚C
Temperature and Dwell Time: 180˚C, 2 seconds Recommended sealing Equipment: * Efly, Kseal, 4s2 ** Wasp, ThermoALPS300/3000, Kube, Flexiseal, Chameleon, REMP (PHS) *** Agilent VII Plateloc, REMP (LHS/SHS)
Clear and thick plastic seal. Sealing surface on inside of roll and is less reflective.
Flexible plastic, difficult to crease, upper surface feels very smooth, sealing side has a slight rough feel. Temperature Range: -80˚C to +80˚C.
Confirming the materials ability to resist high temperatures. Results: Pass
Details: Mass loss of solution evaluated after 30 cycles of 3 step PCR Programme. Equipment: ABI Thermocycler, Precision Balance.
Measuring the force required to push a standardised needle through the material via compression measuring equipment. Results: N/A
Details: 5 tests run using a standardised needle, ensuring that less than 10N is required to pierce the surface & access the wells. Equipment Instron 3343 Tensometer.
Determining the materials optical clarity by measuring the transmission of emissive dye through the material. Results: Pass
Details: Record the light transmission of a sealed microplate using a Flurophore dye stock solution and a microplate reader. Equipment BMG Labtech - FluroStar
Measuring the materials permanence of adhesion & its ability to be removed, via extension measuring equipment. Results: Pass
Details: Cohesive Failure, Adhesive Transfer, Material tear & Successful Peel are measured & recorded after a 180˚ peel test. Equipment Instron 3343 Tensometer.
Low Temperature Seal Test
Confirming the materials ability to resist low temperatures. Results: Pass
Details: Microplates are sealed at specified low temperatures & subjected to a series of tests to substantiate seal integrity. Equipment: Laboratory Cold storage unit.
Evaluating the materials resistance to solvents (DMSO used as an aggressive standard) Results: N/A
Details: Sealed plate is subjected to a high concentration of DMSO for a time period at low temperatures after which seal damage & volume loss are determined. Equipment Laboratory Cold storage unit, DMSO solution.
Plate Types, Sealing Temp. Time Settings
Polypropylene (PP), Polyethylene (PE), Polystyrene (PS), Cyclo Olefin Copolymer (COC)
Details: Temperature and Dwell Time: 175˚C, 2 seconds.
QC testing is carried out to ensure that products are free from nucleases (DNases & RNases) as well as human genomic DNA. Although every effort is made, including cleanroom manufacture, to maintain this level of cleanliness, best laboratory practice with regards to duplicate testing should be followed.